Q: How to solve the problem of crosstalk when using multiple TFmini Plus?
A: Hello, TFmini Plus will not generate crosstalk unless they are installed opposite. If you need to install TFmini Plus opposite under special conditions, you can set the TFmini Plus to a single trigger mode and stagger the measurement time by frequency modulation to avoid LiDAR Interfere with each other.
Q: How can the LiDAR be adjusted to a single test?
A: Hello, achieving a single trigger measurement can be performed in the following ways: Send the command to modify the output frequency, modify the output frequency to 0, and turn on the single trigger measurement switch: 5A 06 03 00 00 63 Send the trigger command: 5A 04 04 63. Each time the command is sent, the TFmini Plus measures it once and outputs the result. Save the configuration instruction: 5A 04 11 6F.
More Common problem here